Our products are designed and manufactured according to standards, including respirator standards such as EN 149、EN 143、NIOSH 42 CFR Part 84、EN 14683、ASTM F3502、GB 2626、GB/T 32610、GB 19083、YY 0469、YY/T 0969, etc. Filter standards such as ISO 16890、EN 1822、ISO 29463、ISO 11155、IEST-RP CC001.5、GB/T 14295、GB/T 6165、GB/T 13554, etc.
From the perspective of users, our products have good practicability and high accuracy. Our customers are distributed in filter media, masks, filters (element), clean rooms and other industries, as well as environmental protection industries with monitoring requirements for air quality. Customers include relevant production enterprises, testing institutions and research institutes at home and abroad.
For medical face masks, we often see these characters such as PFE, BFE and VFE. What is the meaning of these indicators? There is a certain correlation between PFE and BFE. For medical face masks, with NaCl aerosol, flow rate 30L/min, when the PFE is higher than 90.0%, the BFE can be higher than 99.0%. Compared with VFE, PFE, when the PFE reach 91.83%, the VFE can reach more than 98%, even higher than 99%. Test aerosol of BFE and VFE are bacteria and bacteriophage respectively, and biological experiment is a very troublesome and complicated process. For mask manufacturers, these indicators are too complex as routine tests. Therefore, some indirect experiences are often used in practice.
ISO 29463 originates from EN1822, which defines EPA, HEPA and ULPA filters commonly used in industry. While ISO 29463 maintains the classification of EPA, HEPA and ULPA.But replace E10-E12, H13-H14 and U15-U17 with the following 13 filter levels. The ISO 29463 can't replace EN 1822, EN 1822 will continue to be valid.
The reliability of "Scince Purge" testing equipment has been recognized by many customers, especially 1406D, 1406YO, 1802D, 1802YO and other models are favored in the second-hand market. Recently, we have received feedback from the market that due to poor use and maintenance, improper operation of untrained personnel,etc, the following problems have occurred. Including the internal parameters of the equipment are confused, the circuit board is corroded, tetc. In addition, we found that some employees who left our company continued to engage in the after-sales service without our authorization, and resell second-hand "Scince Purge" equipment. The above situation will not exclude the behavior of modifying some internal parameters of the equipment in violation of the principles of our company, which will inevitably lead to the distortion of the test results. Therefore, our company decide to take measures helps the end users to identify and avoid the adverse mpact of such events on users.